Rotary evaporators must have good corrosion resistance and sealing, especially suitable for the extraction of materials that need to avoid metal ion contaminants.
What are the advantages of 5l/2l rotary evaporator?
The advantages of the 5l/2l rotary evaporator are the small structure, compact and efficiency, intuitive work, and no metal ion pollution. 5l/2l rotary evaporator is suitable for laboratory, production pilot, and extraction of valuable materials. Its sealing performance is comparable to foreign machines.
Especially for the concentration of the extract and the distillation of the receiving liquid during chromatographic separation, the reaction product can be separated and purified.
Structure of 2/5 liter rotovap
The other opening of the reflux condenser tube is connected to a receiving flask with a ground mouth. Receive the evaporated organic solvent. When the system is in communication with the atmosphere, the distillation flask and the wetted flask can be removed to transfer the solvent. When the system is in communication with the decompression pump, the system should be in decompression.
When it is finished, the 2/5 liter rotovap should be shut down first, and then the atmosphere should be vented to prevent the distillation flask from falling off during rotation. The function of rotary steaming is to evaporate the solvent, which is more efficient than general methods.
Principle of 2/5 liter rotovap
One is to rely on decompression; the other is to make the solution form a liquid film when rotating, expanding the evaporation area. The following is the method of using the 2/5 liter rotovap and the attention: a small amount of alumite should be added to the installation interface part to avoid the leakage of the interface part and the vacuum degree cannot be increased when vacuuming. Vacuum siliceous oil is required.
Ordinary alumshilin is easy to stick and cannot be disassembled when the vacuum pressure is high! The liquid level in the bottle should be lower than the liquid level heated by the water bath to avoid damage to the bottom shaft! Adjust the angles of the eggplant-shaped bottle, the water bath, and the port of the rotary steamer, and then load the sample into the eggplant-shaped bottle to prevent sample loss and damage to the port of the rotary steamer when the instrument is installed.
After the eggplant-shaped bottle containing the sample is successfully connected to the port of the rotary steamer, the 2/5 liter rotovap button should be activated to check whether the rotation is flexible, and more importantly, whether all or most of the rotary steamed sample is submerged in the liquid surface of the water bath.
Only in this way can the efficiency of rotary steaming be guaranteed. Generally, the sample contained in the eggplant-shaped bottle should not exceed 50%, and the maximum should not exceed 75%, otherwise suction will occur during decompression.
Next, install the container containing the distillate, pay attention to use some alumite when connecting, ensure a good airtight, pass the cooling water, close the vent valve, open the decompression device, and perform the decompression distillation. When the reflux liquid is dripping rather than flowing the best time.
If it is an organic solvent, you need to turn on the freezer in advance to reduce the temperature to a very low level, and during the rotary steaming process, the speed cannot be too fast, at least the evaporation speed cannot be greater than the condensation speed, and the condensate distance at the top of the condensation column judging the position of the suction port.
The situation is often directly heated to a certain temperature, the solution will often bump, and slowly heated to this temperature will often not boil. If the volume of vaporized liquid is large, the liquid inlet tube can be used for continuous injection and high efficiency.