Whether the sealing material is installed at the pipe connection and then connected to ensure that unheated air does not enter the drying room. Check static equipment, pipelines, static sealing points, and high-speed centrifugal spray freeze drying equipment for running, leaking, dripping, leaking, and clogging.
The door and observation window is closed and check for air leaks. Check whether the locks, instruments, valves, etc. of the high-speed centrifugal spray freeze drying equipment are normal. Tighten the pollinator at the bottom of the cylinder body and the bottom of the cyclone separator. The pollinator must be cleaned and dried.
Before installation, check whether the sealing ring has fallen off, and tighten the pollinator before it falls off. Click the start button of the centrifugal fan to check whether the rotation direction of the centrifugal fan is correct.
Fill the oil filling port of the centrifugal nozzle motor to more than half of the observation hole, and then turn on the oil circulation pump. Check whether the regulating butterfly valve at the outlet of the centrifugal fan is open.
Whether the connecting pipe of the feed pump is well connected, and whether the rotation direction of the motor and the pump is correct. The top of the drying room should be covered with the spray head to avoid air leakage. Close the outlet of the feed pump.
Check whether the feed pump is normal. Use and check according to the operating rules of the centrifugal pump. The working principle of the spray freeze drying equipment and the operation method of the spray drying equipment
Check and start the spray freeze drying equipment
Turn on the centrifugal fan and check the operation of the fan equipment: whether there is vibration, abnormal noise, odor, etc. of the high-speed centrifugal spray freeze drying equipment, whether the temperature of the motor and bearing is normal, and whether the anchor screws are loose.
First, turn on the steam heating, turn on the electric heater (electric heating is an auxiliary heating function), and check whether there is electricity leakage (if there is electricity leakage, the control cabinet will automatically trip).
Carry out the cylinder body preheating, the preheating temperature is between 180-220°C. Turn off the fan. Place the centrifugal spray head, turn on the centrifugal spray head, slowly adjust the frequency of the atomizer to a certain frequency (determined according to the actual situation), when the spray head reaches the high speed under the frequency condition, turn on the feed pump and turn on the feed pump The outlet valve.
At the same time, open the pollinator at the bottom of the barrel and the bottom of the cyclone to accept the material. During continuous production, it is not necessary to clean each batch of materials, but it needs to be determined according to the corrosiveness of the materials.
Usually, it does not need to be disassembled when washing the centrifugal nozzle, and the screw pump is used for cleaning. Slowly open the valve of the centrifugal nozzle to increase the amount of feed from small to large, otherwise it will cause wall sticking until it is adjusted to the appropriate requirements to keep the exhaust air temperature constant.
The dried product is collected and placed in the pollinator in the lower part of the tower body and the lower part of the cyclone separator. The pollinator should be replaced before it is full. When the pollinator is replaced, the butterfly valve above must be switched off.